Payment & Shipping Terms:
|Name:||CHT-225A Ultrasonic Concrete Strength Tester||Permissible Sound Time Error:||±0.1μs|
|Sensor Frequency:||50KHz||Rebound Kinetic Energy:||2.207J|
|Rebound Springs Rigidity:||785.0±40.0(N/m)||Size:||205mmX173mmX86mm|
CHT-225A Ultrasonic Concrete Strength Tester
CHT-225A concrete strength tester is used to test compressive strength of concrete by two ways:
rebound and ultrasonic-rebound method. Also, it can be used to test sound time and speed in concrete.
Technical specification for inspection of concrete compressive strength by rebound method(JGJ/T23-2011).
Technical specification for detecting strength of concrete by ultrasonic-rebound combine method(CECS02:2005)
1,Multi-function for testing compressive strength of concrete. It is used to test the compressive strength of concrete by rebound method and ultrasonic-rebound method.
2,Get the compressive strength value of concrete by ultrasonic-rebound method in one instrument. it need two instrument usually, one to test sound speed in concrete, the other to test rebound value.
3,Can save test date. This is convenient for laborer data management.
4,Real USB transmission. Data can be transmit to computer for future process and analysis.
5,Powerful data process and analysis software in windows. Test data can be export to word or excel for future process and analysis.
|Permissible sound time error||±0.1μs|
|Rebound kinetic energy||2.207J|
|Rebound springs Rigidity||785.0±40.0(N/m)|
|Rebound bar impact length||75.0±0.3 mm|
|Rebound value permission error||≤±1|
|Average rebound value on Calibration anvil||80±2|
|Power||6 dry cells,work more than 10 times|
Powerful software in windows is convenient to data process and analysis:Real USB assure transmit data to computer quickly. It also can export data to word and excel for future process, save and print data.
Contact Person: Ms. Rebecca Zhou