Payment & Shipping Terms:
|Name:||Coating Thickness Gauge - Leeb250A/251A/252A|
Coating Thickness Gauge - Leeb250A/251A/252A
Functions & Features
High quality metal probes.
Two measuring methods:continuous and single;
Two working mode: direct and batch;
Switch off automatically or manually.
Magnetic Induction (Fe): Measuring the thickness of Non-magnetic coating on magnetic metal substrate, such as aluminum, chromium, copper, zinc, rubber, paint on the base of steel, iron, alloy and magnetic steel .
Eddy Current (NFe): Measuring the thickness of Non-conductive coating on non-magnetic metal substrate, such as rubber, plastic, paint, oxide on the base of aluminum, copper, zinc, tin.
|Measuring principle||Fe||NFe||Fe or NFe|
H refers to the thickness of testing piece
|Minimum resolution (µm)||0.1μm|
|Min curvature of the min area (mm)||Convex1.5 Concave9|
|Diameter of the min area (mm)||Φ7|
|Critical thickness of substrate (mm)||0.5|
|Memory||200 groups measured data|
|Power supply||AAA Alkaline battery|
Main Machine,5 calibration specimens (48.5m,99.8m,249m,513m
1024m),1 probe & substrate, 1 leather sheath
Contact Person: Ms. Rebecca Zhou